How to use:
1: Put clean and dry stuff in the pouch. (Make sure the length of stuff you put is shorter than 3?4 of that of the pouch. Only in this way, seal can be secured. Or it probably will explode.)
2: Keep the keen-edged stuff award the opposite direction of stripping. (In order to make sure stripping safe.)
3: Seal it up with heat-sealing machine under temperature of 220? to 260? (Keep the operating speed in proportion to the working speed of machine)
4: Put the paper stuff next to the plastic stuff during sterilization.Keep the sterilization going in usual way.
5: Confirm color changing of the indicator during sterilizing. It's pink before sterilization, while it turns brown after.
6: Place the pouch in places of dry, dim and clean.
7: When get to use the instrument in the pouch, you only have to open the pouch as the indication says and just take it out.
8: When try to move, transport and store the sterilizing produts, you have to keep it goes as sanitation standard.
9: Use the stuff sterilized only in the usual way within 60 days.
This bag is only adapt to sterilization by steam. And it's not suitable for dry-heating sterilization exceeding 140?. It should be buried at the specified place.
This item is unable to indicate the effect of sterilization, but only can indicate it has had been sterilized because the coloring matter on the pouch is just for indicating sterilizing process.
Storage: Ambient temperature, 50% relative humidity and dull environment.
Duration: 2 years (under ambien temperature)
Steril duration: 6 month (After sterilization)
Please make sure the bag is in intact conditon before using. Any damage, please don't use it.
This item is one-off product. Don't use it twice.
1. Medical Sterilization pouch
2. It is used to wrap recycling medical instruments for sterilization.
3. Size: 5.5*200mm